Keep current with knowledge base articles from METER Environment
Despite its seeming simplicity, air temperature is one of the most difficult environmental parameters to measure accurately. The current best practice involves housing the air temperature sensor in a radiation shield that is either passively ventilated or actively aspirated. Due to design constraints, the air temperature sensor in the new ATMOS 41 all-in-one weather station cannot be fully shielded from solar radiation.
However, since the ATMOS 41 measures wind speed and solar radiation, both of which are primary factors affecting the accuracy of the air temperature measurement, correction is possible.
The air temperature sensor on the new ATMOS 41 weather station is partially exposed to solar radiation, which may result in large errors in measured air temperature (Tair).
Uncorrected measurements showed errors ranging to 3 °C when compared to measurements made in a state-of-the-art aspirated radiation shield.
Because the ATMOS 41 also measured wind speed and solar radiation, it was possible to use a simple energy balance calculation to correct the Tair measurement. After correction, error decreased to < 0.5 °C and yielded better accuracy than commonly used passive ventilation radiation shields.
The energy balance of the thermometer has been re-arranged below to correct for errors due to solar radiation.
αs= absorptivity of temperature sensor to solar radiation (unitless)
St = total incoming shortwave radiation (W m-2)
cp = specific heat of air (J mol-1 C-1)
k = constant describing boundary layer heat conductance
u = wind speed (m s-1)
d = characteristic dimension of temperature sensor (m)
An Apogee TS-100 aspirated air temperature sensor was chosen as the reference standard for Tair. The ATMOS 41 and Davis instruments air temperature sensor in non-aspirated, louvered radiation shield were co-located with the TS-100. A Davis sensor/radiation shield was included to compare ATMOS 41 performance to a typical Tair measurement. Five-minute averaged data was taken over a five day period of variably cloudy conditions in late summer 2015. αs and k from Equation 1 were used as fitting parameters to minimize error in Tair for the ATMOS 41 correction.
The simple energy balance approach worked well to correct air temperature from a partially radiation exposed sensor.
Uncorrected Tair accuracy from ATMOS 41 is comparable to typical non-aspirated radiation shielded air temperature measurement but showed positive bias from solar radiation effects. Radiation-corrected ATMOS 41 outperformed typical radiation-shielded air temperature measurement and yielded 95% confidence interval of well less than ±0.5 °C accuracy.
(All units °C)
ATMOS 41 uncorrected
ATMOS 41 corrected
Average error (bias)
95% conf interval
Max positive error
Max negative error
Table 1. Summary statistics for air temperature measurements for two sensors under evaluation