Clean ATMOS 41 funnel outside and inside, if needed. Directions here.
Remove the spring from the funnel.
Using a scale/balance, verify that the 20 mL mark on a plastic syringe dispenses close to 20 grams; make a new mark on the syringe if needed.
Place ATMOS 41 on a stand/mount.
Connect ATMOS 41 to a data logger.
Use tweezers to dip one filter paper disc in DI water. Re-wet if needed by dipping the filter in the DI water cup for about two seconds.
Place the pre-wetted filter paper disc at the top of the spring hole.
Use the back of a round pen to push the filter paper to the bottom of the hole. The purpose of the filter paper disc is to regulate the flow rate of water through the funnel.
Poke a very small hole through the middle of the filter paper using a metal wire or fine-point pen/pencil (Figure 1). Exercise caution when poking the filter paper to avoid damaging parts below the funnel.
10. Put a cup/beaker under the spout of the drip counter outlet (Figure 2).
a. A towel can be put around the setup to soak up water spills/splatters
11. Open the data logger software
12. Level the ATMOS 41 and check the X and Y orientation angle (Figure 3). Adjust the level to ±2.0 degrees of dead level (0, 0). If the sensor is more than ±2 degrees of dead level (0, 0), then drops can either be missed by the counter or more drop counts could result if the size of the drops change based on the level.
13. Set the measurement/scan interval to 10 minutes (Figure 3). The ATMOS 41 records the accumulated rainfall depth since the last measurement/scan, so setting a longer measurement/scan interval of 10 minutes is important for checking the drip counter.
14. Close the software program and disconnect the cable adapter from the computer.
15. Pretest the filter paper in the funnel by adding about 5 to 20 mL of water.
a. When adding water to the funnel, use discretion by directing the water to the wall of the funnel, not directly into the hole.
16. Monitor the water flowing out of the spout. You should see discrete drops exit the spout. A good flow rate check is to see if 5 mL of water flows through in less than one minute (5 mL/min), while still observing discrete drops exit the spout. A flow rate of 30 to 40 mL/min could be too fast for checking the precipitation calibration.
a. To know the flow rate of water, repeat the pretest (step 15) and use a known volume of water. Use a stopwatch to time how long it takes a known volume of water to pass through the filter. Divide the volume added by the time in minutes to get flow rate (see appendix).
b. If the flow rate is greater than 30 mL/min, then repeat steps 6 to 9 with a new filter paper (remove the wet filter paper first), and make a smaller hole in the new filter paper.
c. If the flow rate is less than 0.5 mL/min, make the hole slightly bigger.
17. Add a little more than 20 mL of DI water in the syringe. Tap air bubbles up. Squeeze any air bubbles out.
18. Squeeze water out of the syringe until the water is at the 20 mL mark on the syringe. Squeeze this excess water back into the DI water cup/beaker.
19. One minute after the 10-minute data logger interval (eg., 8:01, 8:11, 8:21, etc), gently squeeze 20 mL of DI water from the syringe into the ATMOS 41 funnel with filter paper.
a. Wait 10 minutes for the water to flow through. You should see discrete drops exit the spout.
b. If water flows too slowly, make the opening in the filter paper slightly bigger using a metal wire or fine-point pen/pencil.
20. Repeat steps 17 to 19 three to five times.
21. Open the data logger software, connect, and download the data. Check if the 20 mL volume is close to the expected depth of 2.94 mm.
a. ±5% range is 2.77-3.11 mm
b. ±10% range is 2.65-3.23 mm
NOTE: The precipitation accuracy should fall within 5% for < 50 mm/h rainfall. The more realistic range when testing with a syringe and filter paper is 10%. To check 50 mm/h, add 57 mL of water for a 10-minute interval (~8.37 mm).
22. To check other water depths, change the amount of water in the syringe. Calculate the expected depth of precipitation (see Equation 1). The surface area of the ATMOS 41 funnel is 68.08 cm2.
a. If using 10 mL volume, the expected depth is 1.47 mm.
1. ±5% range is 1.40 to 1.54 mm
2. ±10% range is 1.32 to 1.62 mm
b. If using 50 mL volume, the expected depth is 7.34 mm.
1. ±5% range is 6.98 to 7.71 mm
2. ±10% range is 6.61 to 8.08 mm
23. Use tweezers to remove the filter paper.
24. Insert the spring by pushing firmly down and twisting to get the spring reinstalled.