Frequently Asked Questions
- Soil Moisture Sensor - TEROS 11 - Volumetric Water Content - TEROS 12 - GENERAL - TEROS What tools can I use to measure soil moisture along a soil profile of 40cm depth?
- TEROS 21 - Water potential - Soil Moisture Sensor - TEROS 11 - Volumetric Water Content - TEROS 12 - GENERAL - GS3 - TEROS What are important considerations when thinking about measuring water content and water potential in peatlands (with organic soils)?
Variability of your substrate is a big one. There is a lot of variability in soils as well, but we have better mechanisms to capture and account for variability in mineral soils. Good substrate-to-sensor contact is critical and trickier to accomplish (good installation), but it is achievable. You will most likely require a custom calibration for water content.
- Soil Moisture Sensor - ECH2O 5TE - TEROS 11 - ECH2O 5TM - Volumetric Water Content - TEROS 12 - GENERAL - ECH2O EC-5 - TEROS What is the difference between the FDR method and the capacitance method of VWC measurement? Are they the same?
They are technically not the same. For the strictest technical accuracy, FDR makes a frequency scan and capacitance measures the charge time of a capacitor. From a practical standpoint there is little difference in the performance.
- TEROS 21 - Water potential - Soil Moisture Sensor - TEROS 11 - Volumetric Water Content - TEROS 12 - GENERAL - TEROS Would you agree that with the impact of soil moisture on the atmosphere, measuring water content alone is not enough?
It depends on your specific goals. If you are studying the impact of soil water on atmospheric impact then you would need water potential. There are plenty of cases where water content alone is sufficient if you also have information about your soil.