A recurring core issue
Saturated hydraulic conductivity isn’t an easy measurement to make, mostly because of the lack of a simple-to-use tool. Many soil scientists and engineers resort to cobbling together their own contraptions that are either complicated and finicky, or simple and crude. Neither have proved to be effective setups in terms of accuracy or convenience.
Saturated hydraulic conductivity solved
The KSAT is the only automated setup for taking saturated hydraulic conductivity measurements in the lab. In its simplest form, it’s an instrument that uses both the falling and constant head methods on a soil core. Best of all, it’s completely integrated, so you’re also assured of software-controlled engineering that’s fully tested.
Integration: the key to convenience
Unlike typical contraptions, the KSAT comes with everything you need to make a measurement, meaning you can set it up right out of the box. This type of integration also allows the KSAT to take up minimal bench space. But perhaps its biggest benefit is how it complements the HYPROP. Both the HYPROP and the KSAT can use the same soil core because they share compatible sampling rings. This allows you to take saturated and unsaturated hydraulic conductivity measurements and generate a soil moisture characteristic curve to get a complete picture of a sample’s properties, simplifying both processes.
As the only automated instrument on the market, the KSAT makes measurements a lot more convenient. The easy-to-use software performs all calculations, including temperature corrections based on the viscosity of water. You can also look forward to eliminating the need to time outflow, weigh beakers and make judgement calls, which collectively add up to significant time savings.
A higher degree of accuracy
The KSAT boasts a wide range of measurement conductivities from 5,000 – 0.01 cm/d. Plus it reads and stores data automatically on your computer via USB, so human error is reduced. And because temperature is corrected before a measurement is made, data quality is also dramatically improved for results you can truly rely on.
Superior saturated hydraulic conductivity measurements
Full integration. Simple automation. Improved accuracy. The KSAT finally checks off all the boxes you care about when it comes to measuring saturated hydraulic conductivity in a compact instrument that saves you time, hassle, and worry.
- Removes human error
- Directly calculates Ksat
- Temperature corrections
- Completely integrated package
- Small footprint
- Uses both Constant and Falling Head methods
- Easy-to-use software
- Compatible with HYPROP
- Wide range
- Complies with DIN 19683-9 and DIN 18130-1
|Measureable Ksat values (min.)|
0.01 cm/d (0.004 in/d)
|Measureable Ksat values (max.)|
5000 cm/d (196 in/d)
|Hydraulic conductivity (Ks) of the porous plate|
Ks = 14000 cm/d (5512 in/d)
|Typical statistical inaccuracy at constant environmental parameter and constant flow resistance of the soils|
approx. 2% (in practice 10%)
|Pressure sensor accuracy|
1 Pa (0.01 cm WC or 0.0001 psi)
|Temperature sensor accuracy|
0.2 C (0.4 F)
|Sampling ring (also fits with HYPROP)|
Volume: 250 ml (0.066 gal)
Windows 7 or later, Microsoft Framework 3.5
Have a question or problem? Our support team can help.
We manufacture, test, calibrate, and repair every instrument in house. Our scientists and technicians use the instruments every day in our product testing lab. No matter what your question is, we have someone who can help you answer it.
Phone US: +1 509-332-5600
Phone Europe: +49 89 12 66 52 0